Please verify with an oscilloscope that your trigger pulse is present and that the high level is more than 2 V. Never exceed 5 V! Check the Tech Specs for more details. The PILATUS EXT IN (Trigger Input) is terminated with 50Ω, your trigger source must be able to drive that load.
Under certain circumstance the PILATUS cannot wait indefinitely for the trigger to arrive after arming, causing a "DMA Timeout" error. Please check the User Manual (Chapter 14 Camserver Commands, "ExtMTrigger", "ExtEnable") for details.
Yes! To write e.g. that only the det user gets read and write permissions (600, rw - -) please add the line "umask 077" to the .bashrc in the /home/det directory.
The Pilatus images are seen from the perspective of the source. The left side of the image is the left side of the detector; top side of the image is top side of the detecto if you look into the reflecting mylar foil of the detector.
4. I changed the network settings with "yast", now I get the message: "gethostbyname() : No such file or directory" in the camserver window right after startup. What happend?
The reason is probably a missmatch of the hostnames given in /etc/HOSTNAME with the hostname given in /etc/hosts. Make sure they match and restart after the change.
Make sure they match.
Standard settings (Suse) are:
Turn off the detector. Check if all cables are connected correctly on the detector as well as on the computer. Restart detector and Camserver. If the behavior stays the same please contact email@example.com with the serial number and as many details on how it came to this behavior.
6. After installing a new TVX/Camserver version I get: "dcb_count (2) must be >= NDCB (4) Bad return from dcb_initialize() Camera initialization error -- press to exit"
A TVX/Camserver version that is not matching the detector was installed. For example the TVX/Camserver version 2MF (FAST) is used and the detector is a standard 2M. Get the right TVX/Camserver version for your detector. Contact firstname.lastname@example.org with details like serial number previouse TVX version and the version number you plan to install. Don't forget to restart the Computer after installing.
7. After start up it is impossible to get past the DCB 0 read check in the initialization in the camserver window.
- Please try to restart camserver and tvx.
- If the problem persists, close all programs and turn off the detector and wait for one minute. Turn on the detector. Then try to restart the camserver and tvx.
- If you still get the same behavior, reboot the detector computer, restart the detector and start up again.
If the behavior stays the same after you have tried all above mentions things please contact email@example.com with the serial number and as many details on how it came to this behavior.
You need to edit the file calibration.def located in /home/det/p2_det/config/cam_data. Add the line "Detector p100k0149" after the header. In this example
- p stands for Pilatus,
- 100k for a 100k detector please use the approprite value (100k, 300k, 1m, 2m, 6m) and
- 0149 for the second part of the serial number (e.g. on the sticker on the back of the detector).
The word "Detector" is the same for all. Please fill in the values correctly. If you are in doubts contact us.
Assuming you would like to switch to the tvx-7.G.HH-JJKKLL-2M.tgz version. Then you only need to change the link in the /home/det/p2_det/programs/ directory. Open a shell and enter the following:
- cd /home/det/p2_det/programs/
- ln -snf tvx-7.G.HH-JJKKLL-2M tvx
Reloading the driver:
Close camserver and TVX
Then as root please execute the following:
- cd /home/det/gstar/gsd
As normal user again restart camserver and TVX